Kyoritsu 5202 Digital Lux Meter Operation:
Press “Power” switch to turn the light meter On or Off.
2 Selecting The Appropriate Range
Press “RANGE” switch to select 200,2000,20000 Lux.
3 Offset Adjustment
To obtain the most accurate reading adjust “ZERO ADJ.” with a screwdriver to enable LCD to indicate 000
4 Light receiver Operation
Remove the cover of the Light sensor. Place the Light sensor where the measurement of the light source is to be made.
The meter will then conduct auto testing. Read the testing value after the reading becomes stable.
(*Note: To elevate the accuracy of data please try to hold the Light sensor to face the measuring light source vertically).
5 Data-Hold Operation
The user may hold the present reading and keep it on the LCD display by pressing the “HOLD” button.
When the stored data is no longer needed, one may release the data-hold function by pressing “HOLD” button again.
FRONT PANEL DESCRIPTION:
2.Power On / Off Switch
3.Data Hold Switch
7.Light sensor Cover
Care of the Light sensor
*When the meter is not in use keep the cover of the Light sensor in place to avoid damage to it’s surface.
*Do not touch the Light sensor (opal part) with the bare hands.
*To clean the Light sensor use only a dry flannel fabric or the equivalent Any kind of detergent could damage the Light sensor and should not be used
Kyoritsu 5202 Digital Lux Meter Specifications:
|Ranges||0.1 – 19990Lux|
|200||±4% rdg±5 dgt|
|2000||±4% rdg±5 dgt|
|20000||±5% rdg±4 dgt|
|Current Consumption||2mA approx|
|Response Time||2.5times / sec.|
|0 – 50ºC Below 80% RH|
|Storage Temperature Range||-10ºC – 60ºC|
|30º Less than ±3%
60º Less than ±10%
80º Less than ±30%
|JIS C1609 CLASS A|
|Power Source||6F22 (9V) × 1|
|Dimensions||Meter:148(L) × 71(W) × 36(H)mm
Light Receiving Sensor:85(L)× 67(W) × 32(H)mm