How to use Reed LM-81LX
Zero Offset Adjustment of Light Function Before sarts any light measurements, you should zero the light sensor prior to use in a dark environment. To achieve this, place the light sensor end of the meter under a desktop or flat surface to block any light. Pressing the “Unit/Zero Button” will set set the Lux Meter indication to zero. It is recommended that the zero be checked frequently using the above procedure.
Hold Function To activate the Hold function pressing the “Hold Button”. Now the current value of the light measurement will freeze and “HOLD” symbol will show on the display. To cancel this function, ress again the “Hold button”.
Data Record Start the Data Record function by pressing the “Max./Min. Button” once and the “REC” symbol will be shown on the display. Press the “ Max./Min. Button “ once and the “Max” symbol along with the maximum value will appear on the display. Press the “Max./Min. Button” again, the “Min” symbol along with the minimum value will appear on the display.
REED LM-81LX Specifications
|Measuring Range||0 to 20,000 lux; 0 to 2,000 Fc|
|Resolution||1, 10 lux; 0.1, 1 Fc|
|Accuracy||±5% of reading|
|Sampling Time||1 time/second|
|Power Supply||Single 9V battery (included)|
|Dimensions||156 x 60 x 33mm|