The SEW 6280TB is designed for field or workshop use and can test a wide range of instrumentation. The control panel on the SEW 6280TB is divided into three sections with each function having its own section. To operate the test box, the user simply plugs the power cable of the SEW 6280TB into a standard power outlet and plugs the instrument to be tested into the SEW 6280TB. The image shows a “multifunction” power socket on the instrument but all instruments delivered in Australia will be fitted with the correct 3 pin Australian socket.
For RCD tester verification, the SEW 6280TB can simulate a 10mA, 30mA and 150mA RCD with trip times of 150ms or 30ms (in the case of a 150mA simulation). Loop impedance is checked by plugging the meter being tested into the loop tester section of the test box and operating the meter being tested. The bi-directional switch on the SEW 6280TB allows the operator to measure the local supply loop, and then by changing the switch the test box can simulate a loop impedance value (supply loop + 1 Ohm) to verify the accuracy of the loop impedance tester being checked. Finally the SEW 6280TB can check insulation resistance testers, in both insulation resistance mode as well as continuity mode with a number of pre-set values and test voltages.
The SEW 6280TB Instrument Calibration Check Box is used for checking the integrity and calibration of insulation resistance testers, loop impedance testers and RCD testers, as well as multifunction electrical testers, which may have all three functions in one instrument. The SEW 6280TB comes in a hard case for protection in the field.
SEW 6280TB Check Box Features:
Wide range of operating voltages from AC 210V/50Hz to 250V/50Hz
Calibration check of: RCD Testers, Loop Testers and Insulation Testers
Simulate 3 RCD Test currents and Trip times
Wiring checks: Phase, Neutral, Earth
Supplied with test leads and plugs
Plugs into main supply outlets for power
SEW 6280TB Instrument Calibration Check Box Features:
● For checking AC voltage.
● For checking a RCD tester’s test current & time in mA(ms).
● For checking a loop tester.
● For wiring check (Phase, Neutral, Earth).
● For checking continuity.
● For checking insulation (The maximum is 1.2kV).